Hanna Instruments offers a wide variety of pH electrodes that are designed for many different applications. The type of glass used for sensing pH, bulb shape, body material, type of junction, type of reference and electrolyte used are just some of the design considerations.
The HI1331B uses general purpose (GP) glass, spherical bulb, glass body, single ceramic junction and is refillable with 3.5M KCl + AgCl.
General Purpose Glass Formulation
The measurement of pH at very high temperatures is detrimental to the sensitive glass bulb and will shorten the life of it. A pH electrode with general purpose (GP) glass will have a resistance of 100 megaohms at 25°C while the resistance of LT glass is around 50 megaohms at 25°C. As the temperature of the glass decreases in the sample, the resistance of the LT glass will approach that of GP glass. If using GP glass, the resistance would increase above the optimum range, resulting in increased impedance and ultimately affecting the measurement. The HI1331B is suitable to use with samples that measure from 0 to 70°C.
The spherical bulb is for general purpose use. Other tip shapes include conic for penetration and flat tip for surface measurements.
The glass body is ideal for laboratory use. The glass is resistant to many harsh chemicals and is easily cleaned. The glass body also allows for a fast transfer of heat to the internal reference electrolyte. The mV generated by the reference cell is temperature dependent. The faster the electrode reaches equilibrium, the steadier the reference potential.
Single Ceramic Junction
The outer junction of an electrode, also known as a salt bridge, is necessary component of the electrical circuit. The movement of ions must flow through the junction for a steady reading. The outer reference has a single ceramic frit. The ceramic is a porous material that is easily fused with the glass body and has a similar expansion coefficient. A single ceramic junction has a flow rate of 15-20 μL/hour. Other junction types are available with higher flow rates and made with different materials.
The HI1331B is a refillable probe. Since it is a single junction pH electrode the fill solution is the HI7071 3.5M KCl + AgCl. If using a refillable pH electrode, the fill cap should be removed prior to calibration and measurement. Removing the cap creates positive head pressure in the reference cell allowing for higher flow rate of electrolyte through the outer junction. A higher flow rate will result in a faster and more stable reading.
The HI1331B uses a BNC connector. This type of connector is universal in that it can be used on any pH meter that has the female BNC probe input. Other type of connectors include DIN, screw type, T-type, and 3.5mm to name a few. These types of connectors tend to be proprietary for a particular type of meter and are not interchangeable.
Body Material glass Reference single, Ag/AgCl Junction / Flow Rate ceramic, single / 15-20 μL/h Electrolyte 3.5M KCl + AgCl Range pH: 0 to 13 Max Pressure 0.1 bar Tip Shape spheric (dia: 5 mm) Diameter 8 mm Body Length / Overall Length 210 mm / 265.5 mm Recommended Operating Temperature 0 to 70°C (32 to 158°F) Temperature Sensor no Matching Pin no Amplifier no Digital no Cable coaxial; 1 m (3.3’) Connection BNC Applications specific for flasks Warranty 6 months
Customers also purchased: